Specification of 21L
Horizontal autoclave with an internal volume of 21 L , built in steel and aluminum chemically treated against corrosion sterilization chamber , external finishing with epoxy paint cured at 200 ° C with maximum resistance to temperature and oxidation, 1 cycle automatic sterilizing, manual filling 400 ml , safety valve against overpressure , overtemperature safety thermostat , door latch lever type that prevents the door from opening with the airlock door seal system with special silicone rubber, safety fuse in case of voltage variation and 3 trays for surgical instruments . Warm-up time 30 minutes; sterilization time 10 minutes , drying time 30 minutes
Pressure sterilization : 2.3 kgf / cm ² + / - 0.2 kgf / cm ² .
Working pressure : 1.4 to 2.5 kgf / cm ³
Operating Temperature: 134 ° c .
Width : 375mm
Height : 360mm
Power : 1200 W
Weight: 46 kg
Voltage : 220 V or 110 V
Consumption : 0.65/1.2 kW
NOTE: ohmic value RESISTANCE
- Resistors 127V x 12 Ω.
- Resistance 220V x 42 Ω.
TYPE OF RESISTENCE: External strap
- Fuse protection Internal plate = 2
- 220V-8A-1200W (10A fuse)
- 110/127V- 14A-1200W (15A fuse)
- Frequency = 50/60Hz
- Voltage Range = ± 2% (in tests conducted with a range of up to 10%, there was normal).
- Make sure that the ground wire installed on the outlet is actually connected to "earth". Some electricians install the ground wire in the neutral (which is incorrect)